Current Help System: PC-DMIS Vision (View Core Help)

#### Hit Targets Controls

The controls shown in the Hit Targets tab in the Probe Toolbox let you edit, test, and modify the targets used to measure the feature. The following table describes what these controls do:

Define Target Button | Description
| The Coverage drop-down box allows you to quickly create target sections in order to only measure a sub set of a feature. Limiting coverage can decrease feature execution time. For example, a large feature measured at high magnification may require lots of camera positions to get all the edge points. Selecting 10% coverage would only measure edge points at certain locations around the feature amounting to 10% of its form.
| The Hit Target Color drop-down box specifies that color that will be applied to the feature hit targets. This allows you to differentiate between features, or to ensure visibility on different types of surface.
| The Nominal Color drop-down box specifies that color that will be applied to the feature nominal line. This allows you to differentiate between features, or to ensure visibility on different types of surface.
| The Lock Hit Targets to Part button secures the size, position or rotation of the target.
| The Center Hit Target button centers the target or FOV. What actually moves depends on the status of the Lock Hit Targets to Part button. If first select the Lock Hit Targets to Part button, and then select the Center Hit Targets button, PC-DMIS Vision moves the current FOV to the target. This is only available on DCC Motion machines.

If you deselect the Lock Hit Targets to Part button, and select the Center Hit Targets button, the target moves to the current FOV.

| The Insert New Hit Targets button inserts a new target area. You can then set up different parameters for this specific area of the feature.
| The Delete Hit Targets button will let you delete a previously inserted target from the feature.
| The Reset Hit Target(s) button deletes all of the previously inserted target areas from the feature, leaving the single default target.
| The Test Hit Targets button tests the automatic Target Edge detection for the currently selected target(s). PC-DMIS Vision displays any detected points in the Live View tab of the Graphics Display window.