Current Help System: PC-DMIS Settings Editor (View Core Help)

This term is often used to describe the process of creating instructions which appear in the PC-DMIS Edit window. This can be done through typing in the actual entries, selecting menu command from the menu bars, or by touching measurement points on a physical part with the CMM and pressing the jog box's DONE button or the keyboard's END key.

### Auto Cir Hits

Defines the default number of hits PC-DMIS will take while learning
circles using CAD data. The minimum number of hits that may be entered is three. It only changes the number of hits for circles about to be programmed, not for circles that have already been programmed.

Entry Name: AutoCirHits

Entry Type: Whole Number to specify number of hits. Default value is 4.