## Portable Toolbar

Portable Toolbar for Romer interface

The Portable Toolbar contains several icons that allow you to access various frequently used functions and windows that are helpful when programming or measuring with portable devices.

The available toolbar icons for Romer arms are described in this table:

Icon | Description
| Execute | Executes the measurement process for any currently marked feature(s). See "Execute" in the "Using Advanced File Options" chapter in the PC-DMIS Core documentation.
| Execute Feature | Executes the selected feature. See "Execute Feature" in the "Using Advanced File Options" chapter in the PC-DMIS Core documentation.
| Execute from Cursor | Executes the part program starting at the selected feature and then working down the Edit window. See "Execute From Cursor" in the "Using Advanced File Options" chapter in the PC-DMIS Core documentation.
| Mark | Marks the selected feature with the specified mark color and prepares it for execution. See "Mark" in the "Editing a Part Program" chapter in the PC-DMIS Core documentation.
| Mark All | Marks all features in the Edit window. See "Mark All" in the "Editing a Part Program" chapter.
| Clear Marked | Removes all highlights from currently marked features within the Edit window. See "Clear All" in the "Editing a Part Program" chapter in the PC-DMIS Core documentation.
| Edit Dialog | Accesses the dialog box associated with the currently selected command if one exists.
| Import from CAD File | Imports a part model from any one of the supported file types. You can choose to import a different file type by using the Files of Type list. See "Importing a CAD File" in the "Using Advanced File Options" chapter in the PC-DMIS Core documentation.
| Set Alignment | Links the CAD data to the measured data. See "CAD Equals Part" in the "Creating and Using Alignments" chapter in the PC-DMIS Core documentation.
| Probe Readouts | Displays or hides the Probe Readout window. See "Using the Probe Readout Window" in the "Using Other Windows, Editors, and Tools" chapter in the PC-DMIS Core documentation.
| Large Status Window | Displays or hides the Status window. See "Using the Status Window" in the "Using Other Windows, Editors, and Tools" chapter in the PC-DMIS Core documentation.
| Report Window | Displays or hides the Report window. See "About the Report Window" in the "Reporting Measurement Results" chapter in the PC-DMIS Core documentation.
| Quick Start | Displays or hides the Quick Start interface. See "Using the Quick Start Interface" in the "Using Other Windows, Editors, and Tools" chapter in the PC-DMIS Core documentation.
| Auto Circle | This allows quick access to the Auto Feature dialog box without having all auto feature buttons on the screen. Once the Auto Feature dialog box is open, you can select another auto feature if needed. See "Inserting Auto Features" in the "Creating Auto Features" chapter in the PC-DMIS Core documentation.
| Location Dimension | This displays the Feature Location dialog box, letting you add Location dimension commands into the part program. See "Dimensioning Location" in the "Dimensioning Features" chapter in the PC-DMIS Core documentation.
| Create Views | Saves the current view of the part as a separate VIEWSET command which can later be recalled to display the stored view. See "Working with Viewsets" in the "Inserting Report Commands" chapter in the PC-DMIS Core documentation.
| Update Nominals From CAD - Current | This updates the nominals of the current feature to match those contained in the current CAD model. See "Updating Nominals from CAD" in the "Editing a Part Program" chapter in the PC-DMIS Core documentation.
| Update Nominals From CAD - All | This updates the nominals of all the features to match those contained in the current CAD model. See "Updating Nominals from CAD" in the "Editing a Part Program" chapter in the PC-DMIS Core documentation.
| Reset Measured Values to Nominals - Current | This resets the measured values of the current feature to match its nominal values. See "Resetting Measured Values to Nominals" in the "Editing a Part Program" chapter in the PC-DMIS Core documentation.
| Reset Measured Values to Nominals - All | This resets all the measured values of the current feature to match their respective nominal values. See "Resetting Measured Values to Nominals" in the "Editing a Part Program" chapter in the PC-DMIS Core documentation.
| Probe Compensation | This inserts a PROBECOM/ON command into the Edit window. When set to ON, this compensates for the probe radius for each feature that follows the command. You can set this to OFF if you don't want to apply probe compensation. See these related topics:

- The "Parameter Settings: Probing tab" topic in the "Setting Your Preferences" chapter in the PC-DMIS Core documentation.

- The "Probe Compensation" topic in this documentation.

| Launch Configurator | This launches the PC-DMIS Configurator. The PC-DMIS Configurator is a standalone utility to define different configurations of available and supported machine types and interfaces. These configurations will then appear in your Settings toolbar, letting you switch between defined configurations on the fly. See "Using the PC-DMIS Configurator" topic in the "Setting Your Preferences" chapter in the PC-DMIS Core documentation.