core

### Mark Used

(This item pertains to the Probe Utilities dialog box, accessible through Insert | Hardware Definition | Probe.)

The Mark Used command button is found on the Probe Utilities dialog box.

This option automatically searches the current measurement routine for AB angles. PC-DMIS then adds any AB angles that are found to the current probe configuration.

If you select the Automatically Adjust Probe Head Wrist check box (located on the General tab in the Setup Options dialog box) and click the Mark Used button, PC-DMIS may not select all the probe tips for calibration. For more information about the Automatically Adjust Probe Head Wrist check box, see the "Automatically Adjust Probe Head Wrist" topic in the "Setting your Preferences" chapter.